I’m surprised, too as the process didn’t took nearly as long with the previous version of Revo Scan.
Perhaps you should clean up the Fused or Meshed object in Revo Studio (but I doubt you could get it back into Revo Scan to apply the Texture)
Sadly this didn’t work the last time I tried it. By the way I find this workflow very cumbersome.
No, I haven’t calibrated the scanner after updating Revo Scan. I wanted to do that after finish upgrading the firmware but this didn’t work and I’m still waiting for a response from Revopoint in this regard.
@PopUpTheVolume I got my MINI today and have done some test scans, looking good, I will share results in my own thread soon.
But I just want to check, does the revo calibration software work for you on the MINI?
When I click calibrate the program immeddiately shuts down without error.
I am able to connect to the scanner through the software.
But to quick answer your question , the new calibration version is not ready yet , I was testing the new version with MINI and was able to calibrate it but using different method than POP2 since it have now a new board , the calibration software will be out very soon and I will make video how to calibrate MINI .
Yeah I know what you mean, i just tested it to see what happens when you try without the update because some people have said the program is crashing but I don’t don’t get any crash just cameras disconnected, i know its not going work until the software is updated,
I don’t suspect it needs to calibration but it would be nice to be able to check it
My unit calibration score was 0.2500 it was messed up as the RGB camera was not proper aligned or shifted by the impact while shipping , so I got temporary testing build to fix my MINI RGB .
The old software never crashed on me , I just had the wrong board to calibrate my old mini beta .
The new calibration software will tell you if you need to calibrate or not . If it says NO this means you can’t or need to improve the score .
My MINI score changed from 0.2500 to O.O320 but mostly what improved was the capture of textures . The model scanning without color was the same as before .
That’s interesting I wounder want they changed, and if they did it for a reason, maybe something to do with the RGB camera?
Would be useful if you could get them to fix that, i don’t think my mini had the earlier firmware you had I haven’t updated it
I guess it was the RGB camera that cause some of the latest issues with the software etc… the gain setting had to be adjusted in the software as well to be able to scan colors .
It was all perfect for my months before the RGB … but hey it getting there .
For me personally I care mostly about the 3D scanning results than textures and the scanning results improved above the testing units .
Now I can capture the textures as well and not only with MINI but also better with POP2 than before minus the outdoors scanning in the day times.
Something for something…
However I am glad I could help eliminate the major bugs and help improve the units and software . It is better already as it was planned to be and will keep improving.