That may explain why you can’t get as good details , since after my update I can’t either so things are clear , we need new firmware update fixing this issue and the dev team is on it already investigating .
I will let you know when I get news .
What you mean 0.487
The test was before calibration 0.487 or after ?
If after then is us very bad and you need a new calibration .
Can you please show your scans without color , or scan it without color ?
Your scan is too soft on the edges , can you just scan it at 10 cm distance from the scan and scan just one 360 rotation with the maximum speed of the turntable ?
If possible could you share with me the mesh and fuse point do I could help you more .
Your scan is not optimal and I want to find out why and what is going wrong .
I know you scanning bottom and top at the same time in one session what is not recommend when scanning do find details , if you want max accuracy you can’t do that , you need to scan 2 angles separately and merge the fused point cloud together after .